Autoclaves with various specifications, can be 180-250 the bottle, fuel coal, firewood can be.
Method of use:
(1)the safety valve pressure should be at about 147.12kPa to automatically discharge.
(2)before firing in the pot and fill it with water, sterile bottles lying in the pot to steady, relying on outside of the sterile bottle with lid to keep distance of 110 mm.
(3)cover the Pan should be cleared before the impurity of sludge at the rubber ring and the lid to prevent leaking.
(4)sterilization cannot be in the middle of the ceasefire to cool to avoid cotton in water.
(5)after sterilization, should open the valve, gas discharge in the pan until the pressure inside the cooker at the time of 47.04-98.08 kPa, close vent valve. Valve open, drain the pot of water, then the internal gas net in order to take opening bottles.